Hemocytometer is a microcomputer calculator integrated with counting, analysis and monitoring with ten groups, two places and ten carries.
The number is calculated with ten keys and showing ten groups of numbers and totals at the same time. Press “%” button, the device will display all group data integer digit, press again will display all group data decimal digit( the light is on). It is of accuracy, functional stability, good out shape, clear monitoring,
convenient for use and clinical fit, showing the primary bacterial infection or virus infection. Especially fits for the small and middle size hospitals.
1. Prepare with connected the electricit, turn on the power and “0” is showed on the monitor, then push the “Reset” key, clearing all the data to zero.
2. Then, the instrument will show normal eight data with their eight keys correspondently such as Basophilic(BASO), eosinophilic( EOSIN ), MYELO, JUVEN, Neutrophilic stab granulocyte(STAB), Neutrophilic segmented granulocyte (SEG), lymphocyte(LYMPH) and Monocyte (MONO). If you want to count any data, please push the relative key.
3. The others out of the above-mentioned data, customer can define it by himself with the “9” and “10” key.
4. Each key will sounds “toot” meaning the counting is available and adding “1” in the corresponding calculator while the “Σ” meaning adding “1” for total. If no “toot” is heard, the counting is not working.
5. When the simplex is full will display “99”, will hear a longer time “toot”, stop accumulated , other key will go on accumulating, when the total is accumulated to “990” , the counting of this operation is full. As soon as you hear a longer time “toot” , the counting is not working.
6. While you want to finish this operation, please push the key “%” , after hearing the “toot” ,the computer will change the values into percentage on the monitor the total does not change.. If you want to get a more accurate value, push “%” again, it will show the percentage to decimal. (the light is on), (only when the single channel is accumulated, push
“%” key showing “-” , it means the total calculator showing the single channel value and percentage is 100%.)
7. When the operation is finished, please push “Reset” key, the ten groups and “Σ” return to “0” , then you can start another calculation.
|Maximum count category :|
|Count capacity of every group:|
|Total count capacity:|